Self-diffusion and Brownian motionThe self-diffusion of water molecules follows the Brownian motion. source: https://en.wikipedia.org/wiki/Brownian_motion QuestionWhat is the difference between perfusion and diffusion? Diffusion coefficient, diffusion time, and diffusion distanceFick's first law of diffusion: the substance transfer J through an area is proportional to the gradient of concentration c (i.e. δc/δx): where D is diffusivity, also termed diffusion coefficient, has a unit of mm2/s. The higher the diffusivity, the faster they diffuse. Einstein [1] derived the root mean squared displacement of a self-diffusion as , where D is diffusivity, and d is also termed the diffusion distance. Once should note that diffusivity is more like a velocity measurement (distance over time) not a quantity measurement (quantity over time). High diffusivity implies faster diffusion, not necessarily a large quantity of diffusion. LimitationDiffusivity does not quantify the amount of of diffusion Assumes (1) no structural restriction (2) homogeneity in the diffusion environment. RecitalDiffusion coefficient, a.k.a diffusivity, is a velocity measurement quantifying how fast substance diffuses. Diffusion distance: the root mean squared displacement of free self-diffusion. Question1. What does diffusivity measure? 2. What is the diffusion distance at a diffusion time of 32 ms? (the self-diffusion coefficient of water at the body temperature is 3x10^-3 mm^2/s) Diffusion NMRThe effect of diffusion on the MR signal was mentioned by Hahn in 1950 [2]. Carr and Purcell in 1954 [3] furthered derived the mathematical framework and measured the diffusion coefficient using a Car-Purcell-Meiboom-Gill (CPMG) echo train sequence with background diffusion gradient. It can be applied to study how fast molecules (e.g. water, glycerol) diffuses by themselves, and the measurement is thus called self-diffusion coefficient. A more popular approach, the Stejskal-Tanner sequence, also known as the pulse-gradient spin echo (PGSE) sequence, was developed in 1960's to measure the diffusion coefficient [4](Fig. 1A). The modified version of this sequence is still used today in most of the clinical scanners. The diffusion signals (S) and diffusivity (D) have the following relation. δ: diffusion-encoding gradient duration Δ: the diffusion time g: diffusion-encoding gradient γ: gyromagnetic ratioS0: diffusion signal without diffusion weighting (the b0 signal) The diffusion coefficient can thus be measured by acquiring diffusion signals with difference gradient strength |g|. source: Sosnovik, D. E., Wang, R., Dai, G., Reese, T. G., & Wedeen, V. J. (2009). Diffusion MR tractography of the heart. Journal of Cardiovascular Magnetic Resonance, 11(1), 1-15. [link] *NMR is a physical phenomenon in which atomic nuclei in a magnetic field absorb and re-emit electromagnetic radiation. The resonance frequency depends on the strength of the magnetic field and the magnetic properties of the nuclei.source: https://en.wikipedia.org/wiki/Spin_echo SignificanceThe self-diffusion coefficient can be measured using an NMR experiment. Assumes free diffusion. Recital1. Diffusion sequence uses gradients to encode/sensitize diffusion. 2. Diffusion is presented by an exponential signal attenuation. 3. The diffusion signal attenuation is determined by diffusion coefficient, diffusion time, and the strength/duration of the diffusion sensitization gradient. Questions 1.What is the relation between diffusion time, diffusion encoding gradient, and the diffusion signals? 2.What is the relation between diffusion signals and TE, TR ? (TE = time of echo = between the time of RF excitation and the time at the center of k space. TR = time of repetition = time period to excite the same spins again.) 3.A study uses capillary tubes to create a diffusion phantom, and the diameter of the tube is larger than axon. How would you adjust the diffusion time and diffusion gradient strength? 4.Why does a longer diffusion time require a longer TE in the Stejskal-Tanner PGSE sequence? 5.Are diffusion signal mostly T1- or T2-weighted? Why? (T1 = time of the recovery of the longitudinal magnetization after excitation by an RF pulse. T2 = the spin dephasing time in the transverse plane after RF excitation) 6.Scan A has TE=100 ms whereas scan 2 has TE=130 ms. What is the SNR difference in diffusion MRI at white matter? (white matter T2 = 69 ms, T1 = 1080 ms, Mxy = Mxy0 exp(-TE/T2) ) Diffusion MRI: a combination of diffusion NMR and MRIThe combination of a diffusion sequence and k-space imaging makes diffusion MRI possible. The PGSE sequence developed in 1960s allowed for measuring the diffusivity of free water diffusion using an NMR experiment, and it was yet to be combined with MRI (Lauterbur, 1973) to spatially map the distribution of diffusion coefficient. This idea gave birth to diffusion-weighted imaging at the mid-1980's [5], and Michael Moseley introduced its an important role in the diagnosis of ischemic stroke [6]. Diffusion MRI allows for spatial mapping of the diffusion signals. Eddy current induced by gradient switch can affects the signal readout, phase encoding, and base b0 [7]. The solutions are (1) correcting the partial distortion by a linear transformation, (2) using a bipolar pulse to for diffusion encoding (Fig. 1D). source: http://www.diffusion-imaging.com/2012/03/dti-preprocessing-distortion-correction.html Susceptibility artifact is presented by in-plane distortion and through-plane signal loss. Diffusion gradient encoding increases echo time and result in greater susceptibility artifact. A longer diffusion time also increases echo time. To reduce the artifact: (1) Using a stronger gradient coil to reduce encoding duration. (2) Use a stimulated echo sequence that allows for long diffusion time and short TE. (But SNR is reduced by half in stimulated echo). The artifact can be corrected using FSL TOPUP with two phase encoding acquisitions. source: https://fsl.fmrib.ox.ac.uk/fsl/fslwiki/topup/TopupUsersGuide Fat saturation artifactFat saturation artifact becomes prominent as b-value increases. Fat signals can be greatly reduced using fat suppression. However, diffusion MRI relies on signal attenuation (high b-value, low signal) to quantify diffusion coefficient, but fat does not diffuse. As a result, in high b-value images, signals from water are gone while the fat signals remain. Consequently, it causes errors in diffusion analysis. The solutions are (1) adjust FOV to avoid wrapped around of the fat saturation (not practical for high-resolution imaging), (2) use a better fat saturation sequence. The figure shows the fat saturation artifact in b0 (b) and FA map (c). source: Yeh, P. H., Oakes, T. R., & Riedy, G. (2012). Diffusion tensor imaging and its application to traumatic brain injury: basic principles and recent advances. Correction using FSL EddyThe most popular tool for correcting eddy current and phase distortion artifact is FSL eddy. see https://fsl.fmrib.ox.ac.uk/fsl/fslwiki/eddy eddy includes the following corrections: If no TOPUP result: 1. Register all DWIs to b0 to correct for eddy current artifact and subject movement 2. Register slice to the volume within each DWI (optional) to correct slice-specify eddy current artifact 3. Correct signal dropout in slices If TOPUP result supplied: 1. Restrict registration when registering all DWIs to b0 2. Restrict registration when registering slices to volumes (optional) 3. Correct susceptibility artifact 3. Correct signal dropout in slices Eddy works for single-based diffusion data and does NOT work for non-shell based data (e.g. DSI grid data). Recital 1. Diffusion MRI usually suffers from the susceptibility artifact, fat saturation artifact, and eddy current artifact. 2. The eddy current artifact can be corrected using a software tool (provided that the SNR is good enough) or a set of specially designed diffusion gradients. Questions 1. What is the relation between diffusion gradient strength, diffusion time, and eddy current artifact, susceptibility artifact, fat saturation artifact in diffusion MRI? 2. How to correct for Eddy current distortion? Is there any limitation? 3. How to correct for susceptibility artifact? Is there any limitation? 1. Load an example diffusion MRI data set at https://pitt.box.com/v/course-artifact . Create an SRC in DSI Studio and open it in STEP2 Reconstruction to see the raw DWI. 2. Point out where eddy current distortion, susceptibility artifact, fat saturation artifact are. 3. Compare the b0 with one DWI. Which regions brighten up? Why? 4. Use DSI Studio to do a quality control assessment on DWI data. Reference[1] Einstein, Albert. Investigations on the Theory of the Brownian Movement. Courier Corporation, 1956. (pdf) [2] E L Hahn (1950). Spin echoes. Physical Review, 80:580–594. [3] Carr, H. Y., & Purcell, E. M. (1954). Effects of diffusion on free precession in nuclear magnetic resonance experiments. Physical review, 94(3), 630. [link] [4] Stejskal, E. O., & Tanner, J. E. (1965). Spin diffusion measurements: spin echoes in the presence of a time‐dependent field gradient. The journal of chemical physics, 42(1), 288-292. [link] [5] Le Bihan, D., Breton, E., Lallemand, D., Grenier, P., Cabanis, E., & Laval-Jeantet, M. (1986). MR imaging of intravoxel incoherent motions: application to diffusion and perfusion in neurologic disorders. Radiology, 161(2), 401-407. [link] [6] Moseley, M. E., J. Kucharczyk, J. Mintorovitch, Y. Cohen, J. Kurhanewicz, N. Derugin, H. Asgari, and D. Norman. "Diffusion-weighted MR imaging of acute stroke: correlation with T2-weighted and magnetic susceptibility-enhanced MR imaging in cats." American Journal of Neuroradiology 11, no. 3 (1990): 423-429. (link) [7] Jezzard, P., Barnett, A. S., & Pierpaoli, C. (1998). Characterization of and correction for eddy current artifacts in echo planar diffusion imaging. Magnetic resonance in medicine, 39(5), 801-812. |
Course and workshop >